I have to move an agilent QTOF a few miles and the cost from agilent is alarming. I think i can do this myself if I had the knowhow. Has anyone been in this situation and done it themselves.I know it involves removing the flight tube etc and my main concern is those delicate mirrors etc. Anyone able to share the knowhow and protocol or has seen it disassembled. Any advice cautions etc welcome
Whether you are building your own mass spec, modifying an existing one, or just need help with instrument settings post about it in here.
1 post • Page 1 of 1
Who is online
Users browsing this forum: No registered users and 2 guests