I used integrated tip column before and just switched to separate ones due to special column. Has anyone met this unstable spray issue? I noticed very small bubbles coming out from the column and then get into the tip. I used long column and the pressure is high (600 bar) and this issue is understandable to me. Is there anyone know how to solve it? I tried to use a metal needle and it is better. But the chromatogram peak shape is still not as good as the integrated ones. Any suggestions? Thank you.
Whether you are building your own mass spec, modifying an existing one, or just need help with instrument settings post about it in here.
3 posts • Page 1 of 1
- E. Coli Lysate Member
- Posts: 107
- Joined: Wed Dec 21, 2011 8:22 pm
- Ubiquitin Member
- Posts: 53
- Joined: Thu Dec 15, 2011 7:58 am
Who is online
Users browsing this forum: No registered users and 1 guest