My name is Harsha. I am a PhD student in university of antwerp, belgium.
I connected ESI source to LTQ XL mass spectrometer. To construct the method, I should not to use Sheath gas.
When i turn off sheath gas flow. The instrument (mass spec) is going to stand by mode and an error message pops up saying that sheath gas flow is less than minimum.
Is there a way that the instrument is ONN without sheath gas flow by having an ESI source.
Whether you are building your own mass spec, modifying an existing one, or just need help with instrument settings post about it in here.
1 post • Page 1 of 1
Who is online
Users browsing this forum: No registered users and 0 guests